Main Article Content
Background-Shortages of personal protective equipment (PPE) including N95 filtering facepiece respirators is an urgent concern in the setting of the global COVID-19 pandemic. Decontamination of PPE could be useful to maintain adequate supplies, but there is uncertainty regarding the efficacy of decontamination technologies.
Methods-A modification of the American Society for Testing and Materials standard quantitative carrier disk test method (ASTM E-2197-11) was used to examine the effectiveness of ultraviolet-C (UV-C) light, a high-level disinfection cabinet that generates aerosolized peracetic acid and hydrogen peroxide, and dry heat at 70°C for 30 minutes for decontamination of bacteriophages Phi6 and MS2 and methicillin-resistant Staphylococcus aureus (MRSA) inoculated onto 3 commercial N95 respirators. Three and 6 log10 reductions on N95 respirators were considered effective for decontamination and disinfection, respectively.
Results-UV-C administered as a 1-minute cycle in a UV-C box or a 30-minute cycle by a room decontamination device reduced contamination but did not meet criteria for decontamination of the viruses from all sites for any of the N95s. The high-level disinfection cabinet was effective for decontamination of all the organisms from the N95s and achieved disinfection with 3 disinfection cycles over ~60 minutes. Dry heat at 70°C for 30 minutes was not effective for decontamination of the bacteriophages.Conclusions-UV-C could be useful to reduce contamination on N95 respirators. However, the UV-C technologies studied did not meet our criteria for decontamination under the test conditions used. The high-level disinfection cabinet was effective for decontamination of N95s and met criteria for disinfection with multiple cycles.
Pathogens and Immunity abides by Creative Commons BY 4.0:
This license lets others distribute, remix, tweak, and build upon your work for any lawful purpose, even commercially, as long as they credit you for the original creation. This is the most accommodating of licenses offered. Recommended for maximum dissemination and use of licensed materials. The authors maintain copyright of their materal.
*Due to a template error on our pdfs, articles published from May 20, 2016 to June 24, 2022 incorrectly state the copyright is held by Pathogens and Immunity. Copyright of all articles is held by the authors of each article as noted in the above copyright policy.
1. Tomas ME, Kundrapu S, Thota P, Sunkesula VC, Cadnum JL, Mana TS, Jencson A, O'Donnell M, Zabarsky TF, Hecker MT, Ray AJ, Wilson BM, Donskey CJ. Contamination of the skin and clothing of healthcare personnel during removal of personal protective equipment. JAMA Intern Med 2015;175:1904-10. doi: 10.1001/jamainternmed.2015.4535.
2. Ranney ML, Griffeth V, Jha AK. Critical Supply Shortages - The Need for Ventilators and Personal Protective Equipment during the Covid-19 Pandemic. N Engl J Med 2020 Mar 25. doi: 10.1056/NEJMp2006141.
4. Lore MB, Heimbuch BK, Brown TL, Wander JD, Hinrichs SH. Effectiveness of Three Decontamination Treatments against Influenza Virus Applied to Filtering Facepiece Respirators. Ann Occupat Hyg 2012;56:92–101. https://doi.org/10.1093/annhyg/mer054
5. Viscusi DJ, Bergman MS, Eimer BC, Shaffer RS. Evaluation of Five Decontamination Methods for Filtering Facepiece Respirators. Ann Occup Hyg 2009;53:815–827. https://doi.org/10.1093/annhyg/mep070
6. Heimbuch BK, Wallace WH, Kinney K, Lumley AE, Wu C, Woo M, Wander JD. A pandemic influenza preparedness study: use of energetic methods to decontaminate filtering facepiece respirators contaminated with H1N1 aerosols and droplets. Am J Infect Control 2011;39:e1-e9. doi: 10.1016/j.ajic.2010.07.004.
7. Fisher EM, Shaffer RE. A method to determine the available UV‐C dose for the decontamination of filtering facepiece respirators. J Appl Microbiol 2011;110: 287-295. doi: 10.1111/j.1365-2672.2010.04881.x.
8. Lowe JL, Paladino KD, Farke JD, Boulter K, Cawcutt K, Emodi M, Gibbs S, Hankins R, Hinkle L, Micheels T, Schwedhelm S, Vasa A, Wadman M, Watson S, and Rupp ME. N95 Filtering Facemask Respirator Ultraviolet Germicidal Irradiation (UVGI) Process for Decontamination and Reuse. 2020. https://www.nebraskamed.com/for-providers/covid19
9. Schwartz A, Stiegel M, Greeson N, Vogel A, Thomann W, Brown M, Sempowski GD, Alderman TS, Condreay JP, Burch J, Wolfe C, Smith B, and Lewis S. Decontamination and Reuse of N95 Respirators with Hydrogen Peroxide Vapor to Address Worldwide Personal Protective Equipment Shortages During the SARS-CoV-2 (COVID-19) Pandemic. Applied Biosafety: Journal of ABSA International. 2020. Preprint. https://absa.org/decontamination-and-reuse-of-n95-respirators-with-hydrogen-peroxide-vapor-to-address-worldwide-personal-protective-equipment-shortages-during-the-sars-cov-2-covid-19-pandemic/
10. Lindsley WG, Martin Jr. SB, Thewlis RE, Sarkisian K, Nwoko JO, Mead KR, Noti JD. Effects of Ultraviolet Germicidal Irradiation (UVGI) on N95 Respirator Filtration Performance and Structural Integrity. J Occup Envviron Hyg 2015;12:509-517. DOI: 10.1080/15459624.2015.1018518.
11. 3M Personal Safety Division. Disinfection of Filtering Facepiece Respirators Technical Bulletin. 2020. https://www.3m.com/3M/en_US/worker-health-safety-us/all-stories/full-story-detail/?storyid=8855304f-01cb-4af2-8937-83096cdb4113
12. Mana TS, Sitzlar B, Cadnum JL, Jencson AL, Koganti S, Donskey CJ. Evaluation of an automated room decontamination device using aerosolized peracetic acid. Am J Infect Control 2017;45:327-329. doi: 10.1016/j.ajic.2016.10.006.
13. Kumar JA, Cadnum JL, Jencson AL, Donskey CJ. Efficacy of a multi-purpose high level disinfection cabinet against Candida auris and other health care-associated pathogens. Am J Infection Control 2019 Dec 23. pii: S0196-6553(19)31016-8. doi: 10.1016/j.ajic.2019.11.029.
16. Smolle C, Huss F, Lindblad M, Reischies F, Tano E. Effectiveness of automated ultraviolet-C light for decontamination of textiles inoculated with Enterococcus faecium. Am J Infect Control 2018;98:102-104. doi: 10.1016/j.jhin.2017.07.034.
18. ASTM International, Designation E2197: Standard Quantitative Disk Carrier Test Method for Determining Bactericidal, Virucidal, Fungicidal, Mycobactericidal, and Sporicidal Activities of Chemicals, 2011.
19. Masiuk T, Kadakia P, Wang Z. Development of a Physiologically Relevant Dripping Analytical Method Using Simulated Nasal Mucus for Nasal Spray Formulation Analysis. J Pharmaceut Anal 2016;6:283–291. doi:10.1016/j.jpha.2016.05.003.
20. Rutala WA, Kanamori H, Gergen MF, Sickbert-Bennett EE, Weber DJ. Susceptibility of Candida auris and Candida albicans to 21 germicides used in healthcare facilities. Infect Control Hosp Epidemiol 2019;40:380-382. doi: 10.1017/ice.2019.1.
21. Nerandzic MM, Thota P, Sankar C T, Jencson A, Cadnum JL, Ray AJ, Salata RA, Watkins RR, Donskey CJ. Evaluation of a pulsed xenon ultraviolet disinfection system for reduction of healthcare-associated pathogens in hospital rooms. Infect Control Hosp Epidemiol 2015;36:192-7. doi: 10.1017/ice.2014.36.
22. Cadnum JL, Jencson AL, Gestrich SA, Livingston SH, Karaman BA, Benner KJ, Wilson BM, Donskey CJ. A comparison of the efficacy of multiple ultraviolet light room decontamination devices in a radiology procedure room. Infect Control Hosp Epidemiol 2019;40:158-163. doi: 10.1017/ice.2018.296.